As the tittle says, after I calibrate the sensor by Visual demo software in windows in the first computer(the calibration result is good), I take the USB of FT sensor to connect ubuntu in another computer, and the data is drift. Then, I take USB back to connect visual demo software, the data is the same as in ubuntu. why could this happen? I know the data will drift https://dof.robotiq.com/discussion/1017/ft300-data-drift. But how would the factory use FT sensor long-time in a day? would they calibrate the sensor every two hours during daily use?